Basic Medical Anesthesia Machine
ShortDesc:Multi-function anesthesia machine, with necessary monitoring and alarm functions, to make sure the operation safety, easy, and reliable in performnce.
CE marked, meet EU clinical requirement;
Alloy frame, humanized handle, easily and conveniently move the machine to any place.
Equipped with high precise Anesthesia Vaporizer. The agent output is continuous and stable, and precise. the agent type: Isoflurane, Enflurane;
Adopting stainless pressure gauge, both gas supply and working pressure could be monitored. The pressure range is 0-1MPa, reading scale is 0.05;
With special protection device for Power Socket and Standby Power Socket, so as to avoid any accidents;
With the Light on the top, for clearly view the screen.
1.Mode: Semi~open, Semi~closed or closed system
2.Application: Adult, Pediatric
3.Gas source pressure: 0.25Mpa~0.65MPa
4.Two~tube flow meter ( O2, N2O), range: 0L/min. ~10L/min.O2 and N2O Linkage Device, keep oxygen concentration is not lower than 25%
5.O2 Flush: 25L/min.~75L/min.
6. Lower gas supply alarm: when O2 gas supply is lower, pneumatic audio alarm will arouse
7. Sensor: Rotor type
8. Driven mode: Pneumatically driven electronically controlled
9. Operation method: Keys
10. Respiration Mode: IPPV, Manual,
11. Tidal Volume Setting: 50~1500ml
12. Inspiration / Expiration ratio: 8: 1~1:8
13. Monitoring Tidal Volume: 20~2000mL
14. Monitoring Parameters: Tidal volume, Minute Ventilation, Respiration Mode
15.Alarm Parameter: Tidal Volume Upper/Lower Limit, Minute Ventilation Upper/Lower Limit, Low O2 supply
16. Drawer: 2 Pieces; Writing table: 1 Piece
17. Pressure Gauge: 1 x O2, 1 x N2O
1. Voltage: AC 220V/ 50 Hz or AC 110V/ 60 Hz
2. Input Power: less than 60 VA
3. Max. Input Current: 5A
Fuse on Main Power Supply: 250V 5A φ5X20(F)