Lab Slide Microscope Glass Covers is Made from clear and optically true glass. Covers are useful for holding your specimens flat and in place for observation under a microscope. These high quality cover glasses are uniform in size and free from scratches and striations. Packed in a plastic box for easy handling. Pack of 100 - 18 x 18mm. 0.13mm to 0.17mm thick (#1 Thickness).
·Size: 16mm x 16mm, 18mm x 18mm, 20mm x 20mm
22mm x 22mm, 24mmx24mm
·Thickness: 0.13mm - 0.17mm
1. 7107 Double frosted ends,ground edges, produced by high clear glass sheet,no bubble,no scrape,clear,general glass or supper white glass are available.
2. Slide 7107 can be with 90 corner or 45 corner, frosted end on both sides approx. 20 mm long.
3. Size: 1.0-1.2 mm thick in dimensions of 25.4 x 76.2mm(1" x 3"); 25 x 75mm, 26 x 76mm.
7101: Ground edges
7102: Unground edges
7103: Single concave,ground edges
7104: Double concave,ground edges
7105-1: Single Frosted end,unground edges
7106: Double Frosted ends on one side,ground edges
7107-1: Double frosted ends,unground edges
7108: Double frosted ends on both sides,ground edges
7109: Single Color frosted end on one side,ground edges
7110: Frosted on one side,ground edges
|Designation||Magnification||Dia Of The Field Of View||Focal Length|
|Designation|| Magnification||Numerical Aperture||Working Distance|
| 4X|| 0.10|| 17.912MM|
| 10X|| 0.25|| 2.04MM|
| 40X|| 0.65|| 0.65MM|
| 100X(Oil)|| 1.25|| 0.09MM|
3. TOTAL MAGNIFICATION:
| Objective Total Magnification|
4.Abbe Type Condenser N.A.=1.25
With An Adjustable Iris Diaphragm
5.Coarse Focusing Range: 30MM
With A Coarse Focusing Stopper
6.Fine Focusing Range: 30MM
Fine Focusing Division Interval: 0.002MM
7.Area Of The Stage: 124 X 153MM
Specimen Shifting Range: Transversal 70MM
8.6V20W Built-In Brightness Variable Lamp
Mounting is the base of the microscope. It supports the all weight of the microscope. This mounting has four rubber foots and that makes the instrument stable.
In order to make the observed specimen have enough brightness and make the resolving power of the objectives be used fully, this instrument adopt 6V20W built-in brightness variable lamp.
Turn the iris of the condenser and make the aperture of condenser suitable for the aperture of objective. The condenser is composed of two parts. One is installed on trestle. This trestle can move up and down by rack and gear. The light axis of the condenser must coincide with the light axis of this instrument. When they diverge, adjust 3 crews in the trestle of the condenser. The other is installed on the mounting.
The mechanical stage can make the specimen move transversely and longitudinally.
4.EYEPIECE, OBJECTIVE AND NOSEPIECE:
The microscope imaging system is composed of eyepiece and objective. According to their magnifications, rotate the objectives into the threaded holes of the nosepiece so that when turning the nosepiece, you can change the objectives and get the needed magnification swiftly. The nosepiece adopts precise and advanced structure. When changing the objectives, the center area of the field view always locates within the range of observing and satisfies the focusing requirement. The eyepiece tube that the eyepieces are inserted into is inclined 450 and so observing is comfortable and convenient.
5.COARSE AND FINE FOCUSING EQUIPMENT:
This instrument adopts coaxial coarse and fine focusing equipment. The knobs are located below the stage. So it is easy to operate. Coarse focusing range is 30mm. Fine focusing range is 30mm too. There is knob beside each coarse focusing knob. Right one is used as tightness adjustment for coarse focusing knob. The other is used as coarse focusing stopper. When you see the specimen clearly, you may lock up the stopper. When you lower the stage and raise it again the stage is at the same level. When you do not need the location, you can turn the flange of the stopper and loose it.
1.Rotate the objectives in sequence, according to their magnification, into the thread holes of the nosepiece. Insert the eyepieces into the inclined eyepiece tube.
2.Put the specimen on the stage. Adjust it and make the specimen in the center of the hole of the stage.
3.First use objective 10X, turn the coarse focusing knob and make the objective near specimen and then observe the image through eyepiece and turn the coarse focusing knob down until you can see the specimen image. After that turn the fine focusing knob until you get a sharp image. Because this instrument adopts precise and reliable nosepiece, you can always observe a clear image with different objectives.
4.Take out the lower condenser and put a plan-concave reflecting mirror and you can also observe the specimen image.
5.THE COMPLETE SET OF THE MICROSCOPE:
| 1||Main Body||1SET|
| 2||Eyepiece WF10X, P16X||2PCS APIECE|
| 3||Objective 4X 10X 40X 100X||1PC APIECE|
| 4 ||Reflecting Mirror||1PC|
| 5||Spare Lamp||1PC|
| 6||Spare Fuse||1PC|
| 7||Cedar Oil||1BOTTLE|
| 8||Operation Instrument||1COPY|
1.The microscope is the same as other optical instrument. Need to be kept in cool, dry, dustless and acidless place. The instrument should be covered with dust guard after using.
2.Never disassemble the lens because they are corrected stringently. If there is stain on the lens, it can wipe out by alcohol, but you should pay attention not seep it into the inside of objectives so as not dissolve glues. Dust on the lens can be wiped away by clean brush.
3.Coarse and fine focusing equipment and nosepiece are precisely constructed and should not be dismantled without authorization.
4.When not in use, keep the objectives in the objective box and cover the eyepiece tube with eyepiece tube cover.